Verathon Inc. Recalls GlideScope GVL Video Laryngoscopes
Published: Nov 30, 2012
Healthcare facilities that have GlideScope GVL reusable blades that are being recalled should stop using the blades and contact Verathon Customer Care to obtain a replacement blade, if one has not already been provided. Customers should also contact Customer Care regarding any other questions about the recall. Customer Care can be reached at (800) 331-2313 (U.S. and Canada) or (425) 867-1348 (international), Monday through Friday from 6:00am to 5:00pm PST for the United States and 6:00am to 4:30pm PST outside of the United States. Customers may also email the company at email@example.com.
The recall includes the following models within the specified serial number ranges:
GlideScope GVL 3, 0574-0007: MD10500 to MD112387
GlideScope GVL 4, 0574-0001: LG105000 to LG112758
GlideScope GVL 5, 0574-0030: XL105000 to XL111798
The firm voluntarily recalled the products after learning about design characteristics leading to cracked and broken blades. FDA has been apprised of this action.
The affected GlideScope GVL blades are at risk of developing stress cracks at the blade tip that may not be readily visible during routine inspection prior to intubation. This cracking may eventually cause the blade tip to break and the product to fail. To date, no patient injuries due to this issue have been reported to Verathon.
GlideScope GVL Video Laryngoscopes are used only by healthcare professionals and were distributed directly to hospitals and other healthcare facilities throughout the United States, Canada, and other foreign countries. Affected GlideScope GVL reusable blades can be identified by reference to the serial number engraved on the metal label on the handle of the GlideScope GVL blade.
Verathon has notified its distributors and customers by mail and by telephone or in-person visits and is arranging for return and replacement of all recalled products.
Any adverse reactions experienced with the use of this product and/or quality problems should also be reported to the FDA's MedWatch Program by phone at 1-800-FDA-1088, by Fax at 1-800-FDA-0178, by mail at MedWatch, HF-2, FDA, 5600 Fishers Lane, Rockville, MD 20852-9787, or on the MedWatch website at www.fda.gov/medwatch.
Verathon Media Contacts: Lorraine Silva, Director of Marketing Operations, Verathon Medical, 425.867.1348 (Ext. 5604).
GlideScope, GVL, the GlideScope symbol, Verathon, and the Verathon Torch symbol are registered trademarks of Verathon Inc. © 2012 Verathon Inc.