New Study Investigates the Performance of Masimo SpHb® During Acute Bleeding and After Fluid Resuscitation
Published: Mar 12, 2018
This press release features multimedia. View the full release here: http://www.businesswire.com/news/home/20180312005323/en/
Masimo Radical-7® with SpHb®, PVi®, and RD rainbow SET™ Sensor (Photo: Business Wire)
In the study, Dr. Adel and colleagues sought to investigate the performance of SpHb in different volume statuses and perfusion states. They utilized intraoperative data from 70 patients scheduled for major orthopedic procedures with anticipated major blood loss. SpHb, as well as Masimo PVi® and Pi, were measured using a Masimo Radical-7® Pulse CO-Oximeter®. Lab-Hb was measured using a Beckman Coulter LH 750 analyzer. SpHb and Lab-Hb were recorded at three times – a baseline reading five minutes after endotracheal intubation, after major bleeding, and after fluid resuscitation – resulting in 210 time-matched readings, which were divided into fluid responsive and fluid non-responsive samples and low and high perfusion index samples.
Using Bland-Altman analysis, the researchers found “excellent correlation” between Lab-Hb and SpHb (r=0.938). In addition, they reported “excellent accuracy with moderate levels of agreement,” as noted below, for various sample sets:
|Sample set||Mean bias||Limits of agreement|
|All||0.01||-1.33 to 1.34 g/dL|
|Fluid non-responsive||-0.08||-1.27 to 1.11 g/dL|
|Fluid responsive||0.09||-1.36 to 1.54 g/dL|
|High Pi||0.01||-1.34 to 1.31 g/dL|
|Low Pi||0.04||-1.31 to 1.39 g/dL|
The researchers also noted that polar plot analysis (angular bias of -4°) showed “good trending ability for SpHb as a follow-up monitor.”
The investigators concluded that, “SpHb showed excellent correlation with Lab-Hb in fluid responders, fluid non-responders, low-Pi, and high-Pi states. Despite a favorable mean bias of 0.01 g/dL for SpHb, the relatively wide levels of agreement (-1.3 to 1.3 g/dL) might limit its accuracy. SpHb showed good performance as a trend monitor.”
SpHb is not intended to replace laboratory blood testing. Clinical decisions regarding red blood cell transfusions should be based on the clinician’s judgment, considering among other factors: patient condition, continuous SpHb monitoring, and laboratory diagnostic tests using blood samples.
@MasimoInnovates | #Masimo
- Adel A, Awada W, Abdelhamid B, Omar H, Dayem O, Hasanin A, and Rady A. Accuracy and trending of noninvasive hemoglobin measurement during different volume and perfusion statuses. J Clin Mon. 2018. https://doi.org/10.1007/s10877-018-0101-z.